What are the maintenance techniques for using ultrapure water equipment in small and medium-sized laboratories in winter?
What are the maintenance techniques for using ultrapure water equipment in small and medium-sized laboratories in winter? Author: Water is essential to the laboratory, with the development of the water of the times has been seriously damaged, so laboratories have experienced the benefits of small and medium sized laboratory with ultra-pure water equipment. But as the equipment is used, various problems will also appear. Therefore, the operating conditions of ultrapure water equipment in small and medium-sized laboratories require special attention. Temperature is the main factor affecting the stable operation of ultrapure water equipment in small and medium-sized laboratories. Then how to maintain the ultrapure water equipment in the small and medium-sized laboratory when the temperature drops? 1. When the equipment is in place, fix the bolts that fix the motor, the bolts of the limit switch, and the connecting bolts inside and outside the switchboard to avoid the equipment running have an accident. 2. Ultra-pure water equipment in small and medium-sized laboratories should be installed in a room with relative temperature and humidity to avoid resin degradation at low temperature, humidity corrosion of components, electrical failure, etc. 3. When the equipment is working, do not adjust the relay time. After the power is cut off, the equipment can be kept clean, and the failure of the motor, switchboard and limit switch can be fundamentally avoided. During the operation, water is supplied at the same time, and all water sources are closed when the power is cut off. When the user stops, the user cannot close it in a short time. 4. The raw water quality conditions should meet the requirements of general sodium ion exchange water treatment equipment. For water quality, there is a large amount of sediment, which enters the machine before sedimentation. It is important to keep the equipment clean every day. 5. If the temperature of the equipment is low for more than half a month, the waste water discharge nylon tube is lifted to the exchange column to prevent it from being lost. When not in use for a long time, the residual salt in the salt tank and valve body should be cleaned. 6. The power distribution board and the equipment body are well assembled. If the user is qualified, the switchboard and the main body of the equipment can be separated. The switchboard is suspended in a fixed position, but it must be ensured that the wiring must be correct. 7. The ultra-pure water equipment in small and medium-sized laboratories should be repaired every other year. The upper and lower plates of the salt tank should be painted for protection. Other conditions need to be inspected and repaired.
Contact Person: Ms.Carrie Chan
TEL : (+86)020-39185731
E-MAIL : firstname.lastname@example.org
Whatsapp/Wechat : +86-13751887537
Water Treatment System Add:9A,Building 8,Lianfa Industrial Park , Luopu Street, Panyu District, Guangzhou City, China
Water Filter Cartridge Add:Zhangcun village,Daze town, Xinhui district, Jiangmen City, Guangdong, China
Sign up for our newsletter
Get the latest news and updates from the leaders in water treatment and process separation.